The experimental procedure consisted of a calibration stage and a 3-D measurement stage. In the calibration stage, each parameter of the different sensors, including every light plane, was calibrated. In our system, was designed at the position , and the calibration values of , , , and were 358.92, 360.21, 2.17, and , respectively. The configuration parameters of our projector, such as the focal length, the pixel size, and the resolution of the LCD screen, were known in advance. We chose Eq. 6 as the scheme of multiple light planes in our system. Moreover, the different multistripe projection patterns corresponding to different measurement depths, as well as the line-shifting method, were designed and labeled for accurate 3-D reconstruction.